rthritis (including rheumatoid arthritis) Sports injuries including sprains and strains Soft tissue damage Ulcers and wounds.Tendonitis Surgical wound healing Trigger point massage and myofascial release Wrinkles, anti-aging, skin rejuvenation Acne scars,hypertrophic scars, keloids and healing of burns.
Preparation before operating
1, Check the power supply of the instrument to confirm it is within the specified scope . Plug one end of the power cord into the power input jack on the back side of the host. The other end connects with power supply.
2, According to the treatment needs (probe A is used to make acupuncture point radiation treatment of small infected parts. probe B is used to make direct radiation treatment of extensive infected injured parts.) Insert the power cord plug of A and B probes into corresponding A and B output ports. The power ports of A and B probes may not be conversely connected.
3, Use safety key to turn on the power switch to boot.
1,Short press " Time "and "Power" buttons in the A or B interface to adjust the treatment time according to the treatment needs(10 to 60 minutes cycle is adjustable) and power(I to 5-speed cyele is adjustable).
2,Place the probes on the parts or acupuncture points in need of treatment.
3,Short press "ON/OFF"button to start the treatment .
4,If suspended treatment needed in the course of treatment, short press the"ON/OFF",and short press it again after a pause to continue the treatment.
5,After the end of treatment, with the safety spoon off the power switch,unplug the power cord after the treatment probe in place
Operating The Instrument
1,When it is used for children.the treatment shall be carried out with instructions of the adults. The laser allergic individual should not use product.
2,Do not cover the cooling vent of the equipment.
3,No probe is allowed to directly aim to the eyes.
4,Probe A is applicable to intraluminal parts or acupuncture points.Probe B is applicable to large area of skin tissue or
joints. Do not press the keyboard panel excessively, or the equipment may be damaged.
5,If any abnormal circumstance be found during the course of treatment,the treatment shall be immediately stop and the equipment shall be shut down.Emergency switch can be used in the case of an emergency operation to ensure personal safety.
6,Wait at least 2-3 minutes after shut down the instrument before restart it.
After The Operation
1,Cut off the power
2,Pull the plug from the socket rather then pulling the cable.
3,Clean the probe with soft medical sterilization cotton.
4,Put the probe on the bracket and cover the equipment with a dust cloth(cover).
Q1. Is it manufacturer or trading company?
A: we are a manufacturer specializing in researching, developing and selling collagen tanning bed and the high-tech of spa enterprise. Since it’establishment ,our company is committed to studying, producing and selling the light therapy machine, spa equipment and other beauty products.
Q2: What’s the mini order of your company?
A: The mini order is 1 set / pieces.
Q3: What about the delivery way?
A: we can offer the door to door services by air or by sea, all just depend on your actual request. Such as DHL, UPS,
TNT,FEDEX,EMS… by air; and the sea transportation.
Q4: What’s the delivery time?
A: For ordinary order, the delivery will be done within 7 -15 working days after payment received.
For OEM/ODM service, it depends on the project report issued by distributors.
Q5: What about your after-service?
A: We offer 1 year warranty and 24 hours calling service.
Q6: Do you have any timely technology supports?
A: we have a professional technology supporting team for your timely services. We prepare the technical documents for you, also you can contact us by telephone, webcam, online chat (skype).whatsapp...
Q7: What’s the payment term of your company ?
A: we can accept Western Union ,TT, Paypal, Alibaba trade assurance order.